Non-Ideal Quiet Zone Effects on Compact Range Measurements
Authors: David Wayne, Jeffrey A. Fordham, John McKenna
Publication: EuCAP 2015
Copyright Owner: IEEE
Performance requirements for compact ranges are typically specified as metrics describing the quiet zone's electromagnetic-field quality. The typical metrics are amplitude taper and ripple, phase variation, and cross polarization. Acceptance testing of compact ranges involves field probing of the quiet zone to confirm that these metrics are within their specified limits. It is expected that if the metrics are met, then measurements of an antenna placed within that quiet zone will have acceptably low uncertainty.
Various methods for determining the uncertainty in antenna measurements have been previously developed and presented for far-field and near-field antenna measurements. An uncertainty analysis for a compact range would include, as one of its terms, the quality of the field illuminating on the antenna of interest. In a compact range, the illumination is non-ideal in amplitude, phase and polarization. Error sources such as reflector surface inaccuracies, chamber-induced stray signals, reflector and edge treatment geometry, and instrumentation RF leakage, perturb the illumination from ideal.
This paper will review, in a summary fashion, the equations that estimate the effect of a non-ideal incident electromagnetic field on an antenna. It will calculate the resulting antenna pattern for a candidate antenna and compare it to the ideal antenna pattern thus showing the induced errors. Parametric studies will be presented studying the error effects of varying illumination metrics on the antenna measurement. In addition, measured field probe data from a compact range will also be used with the candidate antennas to investigate induced errors.
The intent is to provide the reader with insight as to how the typical compact range metrics affect the accuracy of an antenna measurement.
You have requested a Reprint of an IEEE Paper
Copyright 2015 IEEE. Reprinted from EuCAP 2015 Conference.
This material is posted here with permission of the IEEE. Such permission of the IEEE does not in any way imply IEEE endorsement of any of NSI-MI Technologies' products or services. Internal or personal use of this material is permitted. However, permission to reprint/republish this material for advertising or promotional purposes or for creating new collective works for resale or redistribution must be obtained from the IEEE by writing to email@example.com.
By choosing to view this document, you agree to all provisions of the copyright laws protecting it